|Ambient Temperature/Humidity||The surrounding ambient temperature must remain below 104°F (40°C). The working temperature is between -20~+65°C. Allowable relative humidity is 20~90% non-condensing.|
|Electrical||INPUT: 120~240 VAC, 50/60 Hz.|
Active PFC Function, PF>0.95.
OUTPUT: Constant-voltage DC output.
Factory preset at 24VDC, adjustable range 20~26.4VDC.
Maximum current load 13ADC.
|EMI Filters||All MRI room EMI filters must be sized as suitable for power supply load (supplied by others).|
|Housing||Die formed marine grade extruded aluminum. Black TGIC polyester powder coat finish – 5-stage pre-treatment; Salt spray test: 1,000 hours.|
|Listings||Power supply is certified to UL1012 and CSA C22.2 No. 107.1 by Intertek Testing Services. Dry Location only.|
|Note||See Kenall luminaire specifications that utilize remote power supplies for use with this unit.|
|Product Location||The MRIPS-312 power supply must be installed outside the shielded MRI environment. It contains ferrous components that can damage or interfere with MRI equipment. A minimum free-air spacing must exist of 3" around all sides and 6" over exhaust fan. Location should not allow excessive dirt/dust to clog ventilation holes and prevent internal air circulation.|
|Warranty||This product is warranted by Kenall Manufacturing to be free of defects in workmanship and materials for a period of one year from the date of invoice. The DC power supply contained in the enclosure carries a five-year warranty from the date of invoice. Warranty is void if all wiring is not completely shielded in grounded conduit.|
|Wiring/Conduit||All power and signal wiring must be completely shielded in grounded conduit. Total length of low-voltage supply wires, at least 18AWG, from power supply to filter to luminaires, should not exceed 50 feet (15m).|